We offer professional and quick repair of any NMR probes. Our long experience helps us to locate the actual source of your problem and distinguish between faults that arise either in console hardware or in the setup of experiment or in actual probe malfunction. We provide a quick inspection and a price estimate that are free of charge except eventual shipping costs. Additonal information is available upon request.
We have access to all necessary test equipment to assess the RF performance of involved components. We also have access to 300, 400 and 500 MHz NMR spectrometers to test and validate performance.
Typical services provided
Arcing problem; bad sensitivity and RF tuning/matching
Repair starts with careful inspection of all RF components incl coils and concludes with replacing faulty components. All used materials have been tested for being non-magnetic and providing no background signal.
Hint: Other possible factors in low NMR signal: low receiver gain at the preamplifier, incorrect sample position, bad shimming, unblanked gradient amplifier, faulty settings for decoupling (or even a frozen sample due to incorrectly set temperature).
Replacing broken glass and ceramics in the probe body
Broken parts in probe body disturb air flow patterns and increase temperature gradients over the sample.They may also cause pure shimming, mechanical vibrations and electric arcing . We always use NMR grade glass and ceramics and test all probe parameters after repair. Substitution of fault-prone glass components in probes by corresponding well-proven and custom-made Kel-F or quartz pieces is available upon request.
Repairing certain gradient coils in diffusion and high-resolution probes
Depending on coil construction, we rewind or substitute either the whole coil or the active shielding section
Repair of MAS probe; arcing and spinning problem
Sample crash due to lost cap or sudden loss of air pressure may seriously damage a MAS probe. We have provided extensive repair of both damaged mechanical (stator) and RF parts.
Repair of probe accessories for temperature control
Heaters and thermocouples are very often broken. We fixed many times both heater, sensor, and mechanical damages.